The technical knowledge of the technical support department of over 20 years experience has set the technical standards for the assembly of the most reliable condenser in the market. Therefore under a Greek signature, we provide in the market an Oxygen condenser with great advantages for the client. The construction quality, the low cost of expendables, the low price combined with the economical service to ensure long-term incomes for your business.
MOBIAKCARE IRENE – Golden Series has conquered a great portion of the market as its reliability is already obvious, as it is traded for 3 years with great reliability. A quick illustration of its main characteristics follows:
- 1 to 5 liters flow
- Air compressor cooler
- Advanced SKF ball bearing compressor
- Consumption 350 Watt
- Oxygen density 93% ± 3%
- Weight 20.7 kg
- Dimensions L 390mmxW 337mmx H 620mm (10cm taller than Airsep VisionAire)
- Standards: CE; ISO13485
- European made Electronic spare parts;
- Noise level 37dB
- Big digital display screen with total operating time indication, current time since the last operation and auto turn on/off programme up to 5 hours.
- Smart auto diagnoses circuit with sound signaling and light indication with problem detection in the digital display screen.
- Large dimensions compressor with 30% more rotating than competitive devices, together with the excellent ball bearing quality with global reputation SKF.
- A smart circuit of consumption control that allows up to 20% economy.
As defined by the technical manual every 3000 hours the input filter should be replaced. The product’s code is 0808542. This filter is situated behind the machine (behind the black cover that holds the cabinet’s filter) and is necessary to be replaced in the defined period of time.
The black cabinet filter with the code 0808254 can be washed frequently by the user every 3-4 days with plenty of water.
Note that the whole maintenance of the Irene does not require any tool or special technological knowledge
A quick guide of the maintenance programme is illustrated below:
- (Mandatory) Every 3000 hours the input filter should be replaced 0808254.
- (Mandatory) Every 5000 hours or from patient to patient the antiantibacterial ter with code 0807319.
- (Mandatory) Every 2 years the 9 Volt battery with code 0142124.should be replaced
- (Mandatory) Every 2 years the cabinet filter (black sponge) with code 0808254 should be replaced.